The COVID-19 IgG/IgM (Whole Blood/Serum/Plasma) Rapid Test Device utilizes lateral flow technology that is used for the qualitative, differential detection of both anti-SARS-CoV-2 IgM and IgG antibodies. In general, antibodies can be detected 1-3 weeks after infection. This test is intended to screen patients for COVID-19. Combining RNA and Antibody tests can significantly raise the sensitivity for detecting COVID-19 in infected individuals.
Coronaviruses are enveloped RNA viruses that are distributed broadly among humans, other mammals and birds that cause respiratory, enteric, hepatic and neurologic diseases. Four viruses - 229E, OC43, NL63 and HKU1 are prevalent and typically cause common cold symptoms in immunocompromised individuals. Three other strains SARS-CoV, MERS-CoV and SARS-CoV-2 (COVID-19) can be transmitted from between non-human vertebrates to humans.
Positive Percent Agreement (PPA): IgG 96.7%; IgM 86.7%; Overall 96.7%
Negative Percent Agreement (NPA): IgG 98.0%; IgM 99.0%; Overall 97.0%
Clinical Agreement with Characterized Samples
Sensitivity: IgG 96.7%; IgM 100%; Combined 100%
Specificity: IgG 97.5%; IgM 100%; Combined 97.5%
Specimen: Whole Blood, Serum, Plasma
Time to Results: 10 minutes
Shelf Life: 24 months from the date of manufacture
1. Remove the test cassette from the sealed foil pouch and use it as soon as possible.
2. Lay device on flat surface and add specimen (see specific instructions for each specimen type below):
a. For Serum or Plasma Specimen: With the plastic dropper provided, draw serum/plasma specimen to exceed the specimen line, as shown in the diagram below. Hold the dropper vertically and transfer drawn serum/plasma specimen into the sample well (S). Immediately add 2 drops (about 80 μL) of sample buffer to the buffer well (B) ensuring that buffer vial tip does not touch the sample. Avoid air bubbles.
b. For Whole Blood Specimen: Hold the plastic dropper vertically and transfer 1 drop of whole blood (about 10 μL) to the sample well (S) of the test device. Immediately add 2 drops (about 80 μL) of sample buffer to the buffer well (B) ensuring that buffer vial tip does not touch the sample. Avoid air bubbles.
3. Wait for the control line (C) to change from blue to a red color. If, after 2 minutes, the sample has not moved across the test window or if blood is still present in the sample well (S), add 1 additional drop of sample buffer to the buffer well (B).
4. The results should be read in 10 minutes. Do not interpret the result after 15 minutes.
Healgen COVID 19 Antibody Test Instruction Video
Best price and on time. What more can you ask for.
Very quick service
Very timely and reliable. Test strips are of good quality and the price is great 👍👍
Fast and easy service
timely service and good prices